NREL has developed a more precise technology for measuring efficiency of concentrating solar cells, enabling the industry to advance.
- [NREL] Post-Deposition Treatment Boosts CIGS Solar Cell Performance
- [NREL] NREL Develops ZnSiP2 for Silicon-Based Tandem Solar Cells
- Researchers set World Record for solar cell efficiency of 34.5%
- [NREL] NREL Study of Fielded PV Systems Demonstrates PV Reliability
- ScienceDaily: New Technique For Growing High-Efficiency Perovskite Solar Cells
Solar researchers have long been unable to reduce an error that occurs during efficiency measurements of triple-absorber, concentrating photovoltaic (CPV) cells one that is caused by too much spectral irradiance from unfiltered, pulsed xenon solar simulators entering into the bottom subcell during testing.
This condition causes an artificial increase in the measured efficiency. The inaccuracy is significant because concentrator systems manufacturers base their photovoltaic technology choices on these measurements, which if imprecise, can misrepresent the technologies’ performance.
Researchers from the National Renewable Energy Laboratory’s (NREL) Photovoltaic Cell and Module Characterization group, in conjunction with Spectrolab, Inc., have developed a solar cell measurement system, the Spectrolab Model 460 Tunable-High Intensity Pulsed Solar Simulator (T-HIPSS), to mitigate the error. Working with Spectrolab, the researchers requested a customized T-HIPSS 460 for use at NREL. With additional hardware and software modifications, including a custom filter, the researchers developed a version of the T-HIPSS 460 that can be correctly adjusted for state-of-the-art multijunction solar cells.
For researchers in the industry who don’t have access to the T-HIPSS 460, NREL has developed an empirical error correction procedure for correcting measurements made with unfiltered simulators. The efficacy of the procedure has been demonstrated by comparing unfiltered efficiency data against data from the new T-HIPSS 460.
By reducing the errors in the efficiency measurements of multijunction solar cells, NREL and Spectrolab are helping concentrator systems companies develop the most efficient—and therefore most cost-effective—devices to manufacture and deploy. This economical approach helps cut the overall costs of producing solar energy and hastens the commercialization of CPV worldwide.
Key Research Results
NREL and Spectrolab have created the world’s most accurate high-efficiency concentrating photovoltaic cell test bed, and developed procedures for correcting data derived from other systems.
NREL significantly improved the accuracy of multijunction solar cell efficiency measurements.
Reducing measurement errors can help concentrator systems companies make efficient technology choices, improving efficiency throughout the CPV industry and facilitating the growth of commercialization.
Technical Contact: Keith Emery, email@example.com
References: Osterwald, C. R.; Wanlass, M. W.; Moriarty, T.; Steiner, M. A.; Emery, K. A. (2014). Effects of Spectral Error in Efficiency Measurements of GaInAs-Based Concentrator Solar Cells. 26 pp.; NREL/TP-5200-60748. Golden, CO: National Renewable Energy Laboratory, 2014: http://www.nrel.gov/docs/fy14osti/60748.pdf. Osterwald, C. R.; Wanlass, M. W.; Moriarty, T.; Steiner, M. A.; Emery, K. A. (2014). “Empirical Procedure to Correct Concentrator Cell Efficiency Measurement Errors Caused by Unfiltered Xenon Flash Solar Simulators.” http://dx.doi.org/10.1109/pvsc.2014.6925466
Latest posts by Jack (see all)
- Top Tips for a Greener Festival Season - August 23, 2017
- Why is it better to choose heating pumps before the other heating means? - August 23, 2017
- Safeguard Your Home’s Electrical Wiring by Hiring the Top Agencies - June 19, 2017